SmarTracks Timing Gates are based on permanent magnet technology and use magnetic fields to time the location and speed of athletes. Due to the magnetic nature of the product we are frequently asked about pacemaker safety. Can somebody with a pacemaker use SmarTracks Timing Gates enabled tracks or would that affect its operation? Should I be concerned to run on a SmarTracks enabled track or field?
Signposts on the sports fields give the following safety instructions to pacemaker wearers and people with other implants:
“Safety notices: This sports facility is equipped with magnetic timing gates, but can be fully used by athletes with pacemakers or other implants for normal sporting activities. It is recommended that athletes with implants maintain a safety distance of 25 cm (approximately 10 inches) between the implant and the individual magnetic points (in the ground). The strength of the magnetic field on the ground corresponds to that of a standard wall magnet. It is recommended not to place other objects sensitive to magnets directly on the magnetic points. In these cases too, a distance of 25 cm, or approximately 10 inches, from the ground is sufficient. For more information, see http://smartracks.run/safety-instructions/.”
While this is sufficient information for most users, others want to understand in more detail what that means.
We are cautious about making broad statements about how pacemakers might behave in permanent magnetic fieldsas created by SmarTracks Timing Gates.We are not medical professionals, so we cannot offer guidance on pacemaker safety.When asked about safe levels of magnetic fields for pacemakers and other medical devices like ICDs (implantable cardioverter defibrillator), we will always advise to consult a physician for this information.
While we find it important to adhere to the safety statement above,
it does little to help you as a pacemaker wearer to understand the level
of risk involved. It is of course not possible for us to understand the
magnetic susceptibility of every pacemaker and ICD ever implanted.
However, we summarize the state of the scientific literature on the
influence of static magnetic fields on the safety and functionality of
cardiac pacemakers and other implants. This overview can help you, as
the wearer of a pacemaker or other implant, to assess the risks
associated with the normal use of SmarTracks Timing Gates.
What field strength is OK?
Several authoritative sources mention 5 Gauss or 10 Gauss
(respectively 0.5 and 1.0 milliTesla = mT) as the maximum magnetic field
in which a pacemaker should operate properly. These are typical values
that regulators require from pacemaker manufacturers.
Since 2003, all European manufacturers have been obliged by the
European standard EN 45502-2-1 to design pacemakers in such a way that
static magnetic fields up to 10 Gauss (1 mT) have no influence on safety
and functionality. 
The EU Working Directive defines the pacemaker limit in static magnetic fields at 5 Gauss (0.5 mT). 
An investigation of the EU Working Directive in 2015 by the Federal
Ministry of Labor and Social Affairs concludes that the safety and
functionality of pacemakers in static magnetic fields is also guaranteed
at a magnetic field strength of up to 7 Gauss (0.7 mT). 
The International Commission on Non-Ionizing Radiation Protection
(ICNIRP) Guidelines state that “Based on scientific knowledge on the
direct effects of static fields on humans, acute exposure of the general
public should not exceed 400 mT (any part of the body). However,
because of potential indirect adverse effects, ICNIRP recognizes that
practical policies need to be implemented to prevent inadvertent harmful
exposure of people with implanted electronic medical devices and
implants containing ferromagnetic materials, and injuries due to flying
ferromagnetic objects, and these considerations can lead to much lower
restriction levels, such as 0.5 mT (IEC 2002).” 
K&J Magnetics has found several sources that hold to the 10 Gauss (1 mT) limit. 
A report on pacemakers and music players from Boston Scientific, one
of the major manufacturers of pacemakers, states that, “[a]s described
in the pacemaker and defibrillator instructions for use, exposure to
strong magnetic fields >10 Gauss (1 mT) may alter implanted device
Medtronic, another major manufacturer of pacemakers and ICDs, after a
request from K&J Magnetics, indicated a 5 Gauss (0.5 mT) limit for
DC magnetic fields.
Since this topic is a critical medical issue, we suggest like
K&J Magnetics deferring to the lowest, most conservative value of 5
Gauss (0.5 mT).
How do SmarTracks Timing Gates work?
SmarTracks Timing Gates In-Ground are based on strong permanent
magnet arrangements buried below the surface of an artificial track,
under an artificial turf field or other surfaces. Each individual gate
consists of two magnetic rods (MCDs) (typically one right and one left
of the individual running lane).
As the MCDs are typically buried perpendicular to the track, looking
at the ground they only represent a small point (around 25mm wide – 1
inch wide). As they are buried nobody can get into direct contact with
magnetic material. Magnetic fields directly at those points on the
ground can be larger than the recommended 5 Gauss (= 0.5 mT) thresholds
mentioned above. These points are comparable to small kitchen magnets or
whiteboard magnets on the ground. As magnetic fields degrade quite
rapidly we have defined a 25cm (10 inch) zone around each of these
individual points. We recommend pacemaker or ICD wearers not to get
their devices closer than 25cm (10 inch) to those points on the ground.
The following images illustrate how MCDs are integrated under the
surface. A vertically integrated MCD sits a minimum of 2.5cm (1 inch)
below ground level, depending on application also significantly deeper.
In rare cases where the surface does not allow vertical integration,
the MCD is integrated horizontally. Horizontally integrated MCDs
represent a rectangular area of around 500mm in length, and 25mm in
width, with the two extremities of the MCD each generating a magnetic
field as described above.
We performed reference measurements to illustrate the drop of the
magnetic field as you get further away from the MCD. As you can see from
the graph below, the magnetic field generated by the MCDs already falls
below the 5 Gauss (= 0.5 mT ) limit indicated above at a 20cm distance,
starting from the top of the MCD itself. For maximum security we
recommend a safety distance of 25cm (10 inch) from the ground up.
As we move further away from the MCD in the ground, the magnetic
field strength continues to drop significantly. The typical field
strength produced by the MCD at 1 meter from the ground up, which is the
equivalent of the height where the sensor typically sits, is below 0.2
Gauss (0.02 mT).
Based on the current state of scientific literature and the results of our reference measurements, we can derive insights into the impact of SmarTracks Timing Gates on the safety and performance of pacemakers in normal use. These insights can help you, as carriers of a pacemaker or other implant, assess the risks associated with the normal use of our Timing Gates.
Normal Use of SmarTracks Timing Gates for Pacemaker Wearers
For any person wearing a pacemaker or similar device in normal use
they would run or walk on a SmarTracks enabled lane through a gate.
Being upright during walking and running results, depending on the
actual body height, in a typical distance of the pacemaker device or ICD
of more than 1 meter from the individual SmarTracks point. In that
distance, the maximum magnetic fields resulting from crossing a
SmarTracks Timing Gates are already typically smaller than 0.1 – 0.2
Gauss (= 0.01 – 0.02 milliTesla) and as such considerably smaller than
the safe values of < 5 Gauss (or 0.5 milliTesla) discussed above. As a
comparison, the earth magnetic field, while it varies according to the
actual geolocation, is generally around 0.48 Gauss (or 0.048 mT) and
bigger than the maximum signal caused by SmarTracks Timing Gates. We
recommend carriers of pacemakers and other implants to uphold a safety
distance of 25 cm from the floor.
 EN 45502-2-1: Active implantable medical devices. Particular
requirements for active implantable medical devices intended to treat
bradyarrhythmia (cardiac pacemakers).